Search found 3 matches
Search found 3 matches • Page 1 of 1
Here's my jig. I put a pot in so I could adjust the level of attenuation between the speaker level and reference and input probe. There's an internal resistor for impedance measurements and I just remove the clip lead to measure impedance. I'd been using a Cakewalk UA-1G USB audio interface for meas...
- Wed Jan 27, 2021 5:33 am
- Forum: Soundeasy Software Questions
- Topic: Feature Request for SoundEasy V27
- Replies: 67
- Views: 5650
How about an "are you sure" prompt on the "Clear Cad" button. It's in the column right next to the Cad Schematic Editor column and I recently hit the clear cad when I wanted the editor.